Physics of capnography
Advances in main-stream sensor technology
Plug & Play inspiration, as modules
Respironics has recently released several new versions of technology. The CAPNOSTAT® 5 is a new solid state mainstream inspiration, as sensor which is small and lightweight. It provides a variety of airway adapters to monitor adult, pediatric and neonatal patients.
In addition they have released LoFlo®, a new Sidestream sensor. The LoFlo technology has a 50cc sampling rate and has a new external sample cell which is attached to the patient interface rather than housed inside the monitor. This design eliminates the need for the respiratory gas sample to be pulled inside the internal infrared measurement bench and eliminates the associated contamination complications as experienced in older sidestream systems. A wide selection of nasal cannulas, traditional, oral and dual O2/inspiration, as, are available which allow easy monitoring of non-intubated patients.
The Capnostat 5 and LoFlo sensors utilize the same electronics which makes them easily interchangeable depending on the type of patient, intubated or non-intubated, you are monitoring.
PHASEIN Medical Tehcnologies, AB, Stockholm, Sweden, introduced the miniaturized mainstream multigas analyzer. The Journal of Clinical Monitoring and Computing (2009; 23(6)355-61) published a performance review of this miniaturized main-stream multigas monitor (MSGM). The unit uses an infrared technique for measurements of carbon dioxide, nitrous oxide, and inhalation agents, and a fuel cell technique for breath-to-breath oxygen monitoring. The performance of this analyzer was compared to a side-stream monitor (Datex-Ohmeda S/5® in a bench study. The response time of the MSGM was 2-4 times shorter than that of the side-stream monitor. There was a lower bias regarding CO2 accuracy for the side-stream system. At a high breathing frequency (60BPM) the side-stream system displayed damped O2 and CO2 trace with amplitude of less than 65% of the main-stream system. Both systems were unaffected by water vapor during a 3 hour test. It would be interesting to compare this system with analyzers using Microstream technology®.
For details on this advanced technology: